Fuses are a common circuit element and are key to protecting from overvoltage and overcurrent situations that can damage important circuits. Fuse markings are typically stamped symbols located on the metal cap of the fuse. Here is a helpful guide to common fuse markings used with instrumentation, taking a common fuse type as an example: … Continued
Many SIGLENT products are RoHS compliant to EU 2015/863. This is also known as RoHS 3 Check the datasheet for the product in question to find more information.
The SDM3045X, SDM3055, and SDM3065X can return bitmap images of their displays by sending the following command: SCDP
Test Current Source: 1mA Max Resolution: 0.01Ω Input Protection: 1000V Input Protection Open-circuit Voltage: <8V Beep Threshold (short-circuit resistance): from 0 Ω to 2000 Ω
Not at this time. Many instruments support communication via sockets (port 5025) and telnet (port 5024) but the port numbers cannot be changed. If you use VISA, you can implement LAN via TCP/IP, which does not need a port number. Here is a listing of current SIGLENT instruments with Socket and Telnet Support.
Simply access the Save/Recall menu for the instrument in question and select Local. The menu will show used/total amounts. Here, the SDG6X has approximately 83 MB and the SDG1X has 85 MB of memory:
Yes. As long as the signal amplitude and frequency does not exceed the input specifications. When measuring ACV using the ACV mode, the DC components will be blocked. When measuring DCV using DCV mode, the AC components will not be blocked, but you can minimize the AC effects by using a slower test speed or … Continued
Per OSHA, a Nationally Recognized Testing Laboratory (NRTL) is an independent third-part laboratory recognized by the United States Occupational Safety and Health Administration (OSHA) to test and certify products to applicable product safety standards. Examples of NRTLs include UL, CSA, and TUV. Many SIGLENT products are TUV certified, which also covers UL and CSA requirements. To … Continued
Product Type Product Series Socket port 5025 (Y/N) Telnet port 5024(Y/N) Digital Oscilloscope SDS5000X Y Y SDS2000X Plus/HD Y Y SDS2000X N N SDS2000X-E Y Y SDS1000X N N SDS1000X-E Y Y SDS1104X-U Y Y SDS1000CML+ N N SDS1000DL+ N N Waveform Generator SDG6000X Y Y SDG2000X Y Y SDG1000X Y Y SDG800 N N … Continued
For convenience, we recommend connecting your instrument to your LAN using a router that can configure DHCP directly and use an internet switch to connect the LAN to the instruments you wish to control. In some cases, you may wish to directly connect your computer to an instrument LAN port without a LAN available. This … Continued