When performing a log sweep, does the power sweep have log steps or is it linear steps with mathematical interpolation? The power sweep uses linear steps. The freq sweep has log or linear steps, depending on the user selection.
Here are the currently supported calibration kits for the SNA5000A series of VNAs Software revision: 184.108.40.206.7R3 SIGLENT Details F503ME N-type, 50 Ohms F503FE N-type, 50 Ohms F603ME 3.5 mm, 50 Ohms F603FE 3.5 mm, 50 Ohms F504MS N-type, 50 Ohms F504FS N-type, 50 Ohms F604MS 3.5 mm, 50 Ohms F604FS 3.5 mm, 50 Ohms … Continued
SMA and 3.5 mm RF connectors are very similar and should be mechanically compatible, but do have some electrical differences that may be important for your application. For best performance, use SMAs with SMAs and 3.5 mm connectors with 3.5 mm connectors. If you question the connection integrity, you can use connector gages to verify … Continued
N-type Impedance 50 Ohms Power 1 W Standard IEC 60169-16 Durability > 2000 uses Torque 1.35 Nm Spanner 19 mm Temperature Range -15 °C ~ +35 °C
Power source calibration is compatible with R&S NRPxxA/AN and Keysight U2000A/B series USB power meters
Not at this time. Many instruments support communication via sockets (port 5025) and telnet (port 5024) but the port numbers cannot be changed. If you use VISA, you can implement LAN via TCP/IP, which does not need a port number. Here is a listing of current SIGLENT instruments with Socket and Telnet Support.
Simply access the Save/Recall menu for the instrument in question and select Local. The menu will show used/total amounts. Here, the SDG6X has approximately 83 MB and the SDG1X has 85 MB of memory:
Product Type Product Series Socket port 5025 (Y/N) Telnet port 5024(Y/N) Digital Oscilloscope SDS5000X Y Y SDS2000X Plus Y Y SDS2000X N N SDS2000X-E Y Y SDS1000X N N SDS1000X-E Y Y SDS1104X-U Y Y SDS1000CML+ N N SDS1000DL+ N N Waveform Generator SDG6000X Y Y SDG2000X Y Y SDG1000X Y Y SDG800 N N … Continued