Two-channel SDGs (SDG1000X, 2000X, and 6000X) can output two separate arbitrary waveforms.
- Load the waveforms into the instrument
- Select the channel you wish to load to and then press Save/Recall
- Select the arbitrary waveform you wish to output for that channel
- Repeat steps 2-3 for the other channel
- Enable outputs
Here is an example of an SDG1032X with two separate arbs: