1-5Vp, 10MHz +/- 100 ppm (Sine preferred) This indicates that the external clock can be “off” from 10 MHz by +/- 100 ppm. All frequency measurements are scaled according to the clock. If the clock is not 10 MHz, then the frequencies indicated on the instrument will also be off.
Product Type Product Series Socket port 5025 (Y/N) Telnet port 5024(Y/N) Digital Oscilloscope SDS5000X Y Y SDS2000X Plus/HD Y Y SDS2000X N N SDS2000X-E Y Y SDS1000X N N SDS1000X-E Y Y SDS1104X-U Y Y SDS1000CML+ N N SDS1000DL+ N N Waveform Generator SDG6000X Y Y SDG2000X Y Y SDG1000X Y Y SDG800 N N … Continued
For convenience, we recommend connecting your instrument to your LAN using a router that can configure DHCP directly and use an internet switch to connect the LAN to the instruments you wish to control. In some cases, you may wish to directly connect your computer to an instrument LAN port without a LAN available. This … Continued
the Insert a FAT32 formatted USB stick into the front panel Press FILE Set Browser to DIR and use the knob to select Local. The local drive is the internal storage of the instrument. U-Disk0 is the USB device. 4. Use knob to highlight the file you wish to copy (or press Operate and “select … Continued
Currently listed datasheet calculation for SSA Frequency Uncertainty is being updated, but as of 08/26/2020, it has not. Here is the new calculation: We can use below new specification: ± (frequency indication × frequency reference uncertainty + 1/2(span/(sweep points-1)) + 10% × resolution bandwidth + marker resolution) For example: The frequency indication is 500 … Continued
The MAC address can be found on the LAN config interface. Path: System–interface–LAN.
The SIGLENT SSA3000X, SSA3000X Plus, and SVA1000X products all have an EMI option that adds a Quasi- Peak detector selection. The Quasi Peak detector is an integrating detector type that will automatically apply to all active traces on the analyzer. To prevent existing traces from being overwritten with the Quasi Peak detector data, please follow … Continued
If you have an antenna with calibration data, chances are it includes a number of columns of data. One column should be the frequency being tested. The others can include free-range performance, gain, and more. For reference, Antenna Factor (or correction factor) is defined as the ratio of the incident Electromagnetic Field to the output voltage … Continued
The SIGLENT USB-GPIB adaptor currently works with the following SIGLENT instruments: Multimeters – SDM3065X/SDM3065X-SC SDM3055/SDM3055-SC SDM3045X Function/Arbitrary waveform generators – SDG6000X Series SDG5000 Series SDG2000X Series SDG1000X Series SDG1000 Series RF Generators – SSG3000X/-IQE Series SSG5000X/-V Series Spectrum & vector network analyzers – SSA3000X Series SSA3000X Plus Series SVA1000X Series Oscilloscopes … Continued
EasySpectrum is free software used to control and collect data from SIGLENT SSA and SVA spectrum analyzer products. State files are used by the SIGLENT SSA and SVA instruments to recall or save specific instrument configuration settings. By storing and recalling states, you can quickly configure the analyzer for different tasks. To use State files … Continued