The SIGLENT SDS2000X Plus and SDS2000X HD series of oscilloscopes have a 500 MHz bandwidth upgrade option (Click SDS2XP 500 MHz BW or SDS2000X HD 500 MHz BW to find out more). 500MHz bandwidth is available in the following cases: Only one active channel (C1 or C2 or C3 or C4) Two channels (Either C1 … Continued
A number of SIGLENT products feature an embedded program that eases the conversion of binary data to CSV. Click to download the Binary-to-CSV program.
Here is a list of the current products that support Bode plotting and the maximum number of points available per sweep: SDS1xx4X-E/SDS2X-E Series: 500 points (Linear)/1043/decade (Log) SDS2000X Plus/SDS5000X Series:500 points (Linear)/1024/decade (Log) SDS6000A Series: 751 points (Linear)/1576/decade (Log)
Here is a quick method for estimating the lowest current measurement that is possible with an oscilloscope and current probe. This is especially useful when the datasheet for a probe may not show the minimum current that can be measured or it may be missing the noise floor value. For this note, we will be … Continued
Here is a table showing the various memory depths used in the Bode Plot feature Product series Max memory depth SDS1xx4X-E 140k SDS2000X-E 140k SDS2000X Plus/HD 100k SDS5000X 125k
The minimum start frequency is set to 10 Hz because of the extended acquisition time required to collect enough data to ensure a useful level of measurement accuracy. If the low limit was 1 Hz, the scope would need to test 10 cycles (to maintain specified accuracy), so the time base will be 10 s/div. … Continued
Press Utility: Press Options: Items listed as Temporary are trials/demos and will list the number of activations/trials left. Items listed with Permanent have been activated permanently.
USB device: USB 2.0, HS (high speed) 480Mbps LAN:100Mbps
Product Type Product Series Socket port 5025 (Y/N) Telnet port 5024(Y/N) Digital Oscilloscope SDS5000X Y Y SDS2000X Plus/HD Y Y SDS2000X N N SDS2000X-E Y Y SDS1000X N N SDS1000X-E Y Y SDS1104X-U Y Y SDS1000CML+ N N SDS1000DL+ N N Waveform Generator SDG6000X Y Y SDG2000X Y Y SDG1000X Y Y SDG800 N N … Continued
For convenience, we recommend connecting your instrument to your LAN using a router that can configure DHCP directly and use an internet switch to connect the LAN to the instruments you wish to control. In some cases, you may wish to directly connect your computer to an instrument LAN port without a LAN available. This … Continued