Here are the typically expected noise floor numbers for the SDS6000A Oscilloscope: Note: Uses StdDev measurement @ 5 GSa/s, 10 MPts/ch
Fuses are a common circuit element and are key to protecting from overvoltage and overcurrent situations that can damage important circuits. Fuse markings are typically stamped symbols located on the metal cap of the fuse. Here is a helpful guide to common fuse markings used with instrumentation, taking a common fuse type as an example: … Continued
Here is a cross-reference of SIGLENT oscilloscope serial decoding features compared to the Tektronics MDO3 options: Tek MDO3MSO: 16 channel logic analyzer function. The SIGLENT equivalent is optional license SDSxxxx-16LA, where xxxx represents the model (SDS1000X-E, SDS2000X-E, SDS2000X Plus, SDS5000X, or SDS6000A series of oscilloscopes) and hardware (SPL2016 for SDS2000X Plus, SDS5000X, and SDS6000A series/SLA1016 … Continued
A number of SIGLENT products feature an embedded program that eases the conversion of binary data to CSV. Click to download the Binary-to-CSV program.
Here is a list of the current products that support Bode plotting and the maximum number of points available per sweep: SDS1xx4X-E/SDS2X-E Series: 500 points (Linear)/1043/decade (Log) SDS2000X Plus/SDS5000X Series:500 points (Linear)/1024/decade (Log) SDS6000A Series: 751 points (Linear)/1576/decade (Log)
The SP3050A 500 MHz probes for the SDS6A are adjusted to best match the front end of the SDS6A series of scopes. They have a different response from the same models that are included with other SIGLENT scopes, including the SDS5X. The label is used to distinguish between the two types of SP3050A probes.
Here is a quick method for estimating the lowest current measurement that is possible with an oscilloscope and current probe. This is especially useful when the datasheet for a probe may not show the minimum current that can be measured or it may be missing the noise floor value. For this note, we will be … Continued
The minimum start frequency is set to 10 Hz because of the extended acquisition time required to collect enough data to ensure a useful level of measurement accuracy. If the low limit was 1 Hz, the scope would need to test 10 cycles (to maintain specified accuracy), so the time base will be 10 s/div. … Continued
For convenience, we recommend connecting your instrument to your LAN using a router that can configure DHCP directly and use an internet switch to connect the LAN to the instruments you wish to control. In some cases, you may wish to directly connect your computer to an instrument LAN port without a LAN available. This … Continued
Due to channel skew, Bode plotting is limited to 120 MHz for the following products: SDS1104X-E SDS1204X-E SDS2000X-E Series SDS2000X Plus Series SDS2000X HD SDS5000X Series SDS6000A