Simply access the Save/Recall menu for the instrument in question and select Local. The menu will show used/total amounts. Here, the SDG6X has approximately 83 MB and the SDG1X has 85 MB of memory:
Siglent analyzers (SSA/SSA plus/SSA -R and SVA series) are now compatible with the latest version of EMCView software from Tekbox: EMCview PC software for EMC pre-compliance testing
Per OSHA, a Nationally Recognized Testing Laboratory (NRTL) is an independent third-part laboratory recognized by the United States Occupational Safety and Health Administration (OSHA) to test and certify products to applicable product safety standards. Examples of NRTLs include UL, CSA, and TUV. Many SIGLENT products are TUV certified, which also covers UL and CSA requirements. To … Continued
Yes. NOTE: This is not currently available on the SSA3000X series. The quality factor of a filter is defined as: Q = Center frequency/Bandwidth From the SSA3000X Plus, X-R, SVA1000X Series Manual: You can find the automatic Q measurement in the Measurement function menu. Press the Maker Fn button > Select N dB BW … Continued
1-5Vp, 10MHz +/- 100 ppm (Sine preferred) This indicates that the external clock can be “off” from 10 MHz by +/- 100 ppm. All frequency measurements are scaled according to the clock. If the clock is not 10 MHz, then the frequencies indicated on the instrument will also be off.
Product Type Product Series Socket port 5025 (Y/N) Telnet port 5024(Y/N) Digital Oscilloscope SDS5000X Y Y SDS2000X Plus/HD Y Y SDS2000X N N SDS2000X-E Y Y SDS1000X N N SDS1000X-E Y Y SDS1104X-U Y Y SDS1000CML+ N N SDS1000DL+ N N Waveform Generator SDG6000X Y Y SDG2000X Y Y SDG1000X Y Y SDG800 N N … Continued
For convenience, we recommend connecting your instrument to your LAN using a router that can configure DHCP directly and use an internet switch to connect the LAN to the instruments you wish to control. In some cases, you may wish to directly connect your computer to an instrument LAN port without a LAN available. This … Continued
the Insert a FAT32 formatted USB stick into the front panel Press FILE Set Browser to DIR and use the knob to select Local. The local drive is the internal storage of the instrument. U-Disk0 is the USB device. 4. Use knob to highlight the file you wish to copy (or press Operate and “select … Continued
Currently listed datasheet calculation for SSA Frequency Uncertainty is being updated, but as of 08/26/2020, it has not. Here is the new calculation: We can use below new specification: ± (frequency indication × frequency reference uncertainty + 1/2(span/(sweep points-1)) + 10% × resolution bandwidth + marker resolution) For example: The frequency indication is 500 … Continued
The MAC address can be found on the LAN config interface. Path: System–interface–LAN.